This paper is only available as a PDF. To read, Please Download here.
We measured femoral intramedullary pressures and applied axial thrust force generated in vitro during reaming with the AO and Zimmer systems. Six pairs of cadaver femora were instrumented with pressure taps midshaft and in the distal diaphysis, a load cell distally to measure force, and a displacement transducer to monitor reamer position. Following initial hand reaming, intramedullary power reaming was conducted utilizing a 9-mm reamer initially, with subsequent increases in steps of 0.5 mm. All femora were maintained at 37°C and albumin was used to maintain a fluid-filled canal. The highest pressures consistently occurred during initial power reaming, with peak pressures ranging from 270 to 1500 mmHg amongst femora with the AO system. No significant differences were found in the peak pressures generated for the two systems (P =0.10). The pressure measurements at the two locations in the femur were consistently similar, indicating that pressures are continuous throughout this aspect of the femur. The pressures were not correlated with instantaneous applied axial thrust (R2 = 0.191), and this could be attributed chiefly to the additional friction force of cutting. While pressure generation in the medullary canal upon reaming is likely governed by the rate of clearance of canal content, this is a highly variable response produced by characteristics of the femur which are still not fully understood.
To read this article in full you will need to make a payment
Purchase one-time access:Academic & Personal: 24 hour online accessCorporate R&D Professionals: 24 hour online access
One-time access price info
- For academic or personal research use, select 'Academic and Personal'
- For corporate R&D use, select 'Corporate R&D Professionals'
Subscribe:Subscribe to Injury
Already a print subscriber? Claim online access
Already an online subscriber? Sign in
Register: Create an account
Institutional Access: Sign in to ScienceDirect
- Effect of Pressure on Intravasation of Fat from the Bone Marrow Cavity.Surg Gynecol Obstet. 1972; 134: 21
- Pulmonary damage after intramedullary femoral nailing in traumatized sheep — is there an effect from different nailing methods.J Trauma. 1992; 33: 574
- Primary intramedullary fixation in polytrauma patients with associated lung contusion — a cause of posttraumatic ARDS?.J Trauma. 1993; 34: 540
- Influences of different methods of intramedullary femoral nailing on lung function in patients with multiple trauma.J Trauma. 1993; 35: 709
- Pathogenesis and clinical relevance of bone marrow embolism in intramedullary nailing—demonstrated by intraoperative echocardiography.Injury. 1993; 3 (Suppl.): 73
- Fat embolism precipitated by reaming of the femoral canal during revision of a total knee replacement.J Bone Joint Surg [Am]. 1994; 76A: 899
- Intramedullary pressure in reamed and unreamed nailing of the femur and tibia—an in vitro study in intact, human bones.Injury. 1993; 3 (Suppl.): 56
- Effect of flexible drive diameter and reamer design on the increase of pressure in the medullary cavity during reaming.Injury. 1993; 3 (Suppl.): 40
- Intramedullary pressure, strain on the diaphysis and increase in cortical temperature when reaming the femoral medullary cavity — a comparison of blunt and sharp reamers.Injury. 1993; 3 (Suppl.): 22
- Influence of compression force on the intramedullary pressure development in reaming of the femoral medullary cavity.Injury. 1993; 3 (Suppl.): 36
- Influence of the reamer shape on intraosseus pressure during closed intramedullary nailing of the unbroken femur: A preliminary report.Injury. 1993; 3 (Suppl.): 48
- Measurements of intramedullary pressure in an animal experiment and propositions to reduce the pressure increase.Injury. 1993; 3 (Suppl.): 7
- Fat release after femur nailing in the dog.J Trauma. 1983; 23: 322
- Does the reamer type influence the degree of lung dysfunction after femoral nailing following severe trauma? An animal study.J Orthop Trauma. 1994; 8: 300
Accepted: April 4, 1995
© 1995 Published by Elsevier Inc.